Intecont Tersus – Free download as PDF File .pdf), Text File .txt) or read for weighing tasks better than % Manual and/or automatic zero setting Full. The INTECONT Tersus weighing elec- tronics is specially tion module, INTECONT Tersus optimally fits into any Manual and/or automatic zero setting. processes, the INTECONT Tersus meas- uring, control, and supervisory module, INTECONT Tersus optimally fits into any Manual and automatic zero setting.
|Published (Last):||4 September 2007|
|PDF File Size:||17.58 Mb|
|ePub File Size:||12.39 Mb|
|Price:||Free* [*Free Regsitration Required]|
That substantially boosts the short-term accuracy of the scale.
Manual Intecont Tersus
Hbc Radiomatic Micron 5 User Manual? A split mea-suring table can also be used in place of the weighing idlers.
The fact that the weighing system always has the same working point means it can achieve a high level of measuring accuracy. Memory A P Relevant parameters Behavior gersus alarm Triggered: When it reaches the batch setpoint, the conveyor belt and material prefeeder are turned off and the feeding process is ended.
Electric W1 P Maintenance Feeder Run Overall duty cycle of the belt conveyor and the device. Always have qualified personnel ensure that the components are neither damaged nor defective. Using weighing modules with leaf spring parallel guidance, lever ratio is always 1.
Connection diagram for voltage supply The list of all possible event messages and the current hardware allocation is printed out as an appendix. Bvhgb Instruction Manual Intecont Documents.
Return to the normal display, or abort a function.
Parameter block Rated Data: Select the desired function of the second level. DANGER Electric shock from live components Danger of life from electric shocks Take all possible precautions tersks ensure safety before commencing work on live components.
Analyze any event messages that may appear and rectify the fault. Volumetric Synchronous In contrast to volumetric operation, it is not the rated belt load that is used for calculating the set belt speed.
Change back to the first level. In other respects the protection msnual of the tesus must be observed. Preselection batching mode Preselecting the batching mode ingecont batching possible, although intecong does not initiate a batch. The external absolute setpoints of the fieldbus FB or analog output AI can be eva-luated relatively with the keyboard. Attach the calibration weight at the place provided. Access to some functions can be configured in the Dialog Behaviour block.
The effective load is always the applied weight QP. If the optional activation is required on devices which have already been delivered, the data is transferred to the Schenck process, and the generated activation code is reported back. Ontecont This signal word indicates a danger that can immediately cause the most severe injuries up to and including death. The measuring load will always be calculated and displayed.
The second line of the display is reserved for event information. The internal clock will run for a further approx. The parameters values will be converted automatically if the dimensions are changed. Call up the function block The previously selected function will be displayed if no event is pending.
It is output through contact X Subject to change without prior notice. Namur Error Tacho A P Belt Skew A P This starts with the colour LCD featuring high resolution, event marking and input assistance.
Manual Intecont Tersus
intecnot The inclination of the belt may only be entered for systems in which the load cells are arranged perpendicular to the belt. Feeder is no longer operable. In parameters C 08, not the actual weight but the weight affecting the weighing platform must be entered.
Instead, the measured belt load is used before changing over.
The meas-ured feed rate or actual belt load value is not taken into consideration.