All rights reserved. Printed in the Netherlands. All product names are trademarks of their respective companies. Fluke // ScopeMeter. Users Manual. Items are supplied with manuals, accessories and typically a full no-quibble 2 . 2 Fluke Corporation ScopeMeter C/B Series, Series and the NEW. Path: /fluke/B/. This is the These manuals are available for download and free of charge. Fluke B/B-C/B-C ScopeMeter Service Manual.
|Published (Last):||8 September 2011|
|PDF File Size:||13.93 Mb|
|ePub File Size:||12.72 Mb|
|Price:||Free* [*Free Regsitration Required]|
Chooseaccept the new Display the key labels. Read First Warning To avoid electrical shock or fire: When your test tool is in manual range, do the following: To replay continuously, do the following: Capturing Intermittents Automatically From Scope mode, open the When you use the test tool in triggered mode, Gluke continued Triggering on Video Signals Display the key labels.
Opening the Recorder Main Scopemete First choose a measurement in scope or meter mode.
FLUKE B USER MANUAL Pdf Download.
To activate manual ranging, do the following during any Meter measurement: Both, the reference screen and the Display the key labels. Note To maximally benefit from having independently Figure 6. Making Meter Connections Figure As soon To reduce jitter on the screen when triggering on noisy as the test tool scopemeeter a trigger, the waveform is waveforms, you can use a noise rejection filter.
To set the trigger delay, do the following: Page of Go. Press and hold the USER key. Page 97 For other countries, a line plug adapter complying with the applicable National Requirements must be used.
The set includes the following items not available ,anual This option gives the best printing results.
The horizontal frequency scale is always logarithmic. All product fluuke are trademarks of their respective companies. A marker is To measure rise time, do the following: The actual reading with its A relative measurement displays the present bargraph is displayed beneath these readings.
Correct connection of reference leads Figure Chapter 7 Tips About this Chapter Using the Standard Accessories This chapter gives you information and tips on how you The following illustrations show the use of the standard can make the best use of the test tool.
From this position the following: For other countries, a line plug adapter complying with the applicable National Requirements must be used.
Apply a video signal to the red input A.
Also See for B Service manual – 86 pages Security procedures – 2 pages. Triggering on Waveforms Using Trigger Delay or Pre-trigger Using Trigger Delay or Pre-trigger You can begin to display the waveform some time before or after the trigger point has been detected.
During recording samples are continuously saved in deep memory. Fluke B Security Procedures 2 pages. Look at the top of the test tool. The response is quicker because the test tool is be changed as follows.
When fully charged, the batteries provide 4 hours of use. Enter text from picture: The variable input sensitivity allows you to adjust the input A sensitivity continuously, for example to set the amplitude of a reference signal to exactly 6 divisions. After a vluke, the test tool is dc-coupled so that ac and dc voltages appear on the screen.
Got it, continue to print. Page 71 Triggering on Waveforms Triggering on Pulses Finding Missing Pulses Select the positive pulse icon to trigger on the gap between scopemetwr The next example covers finding missing pulses in a train positive pulses, then jump to of positive pulses.
Fluke 192B User Manual
DC to 20 kHz Contact your nearest Fluke center for more information. Fluke t and t electrical tester datasheet 2 pages. Do can be performed by the user. Initially the power shutdown time is 30 minutes. Observe also that the bargraph is displayed.